Refraktometer Kopi dengan rentang pengukuran
Kadar Brix Kopi 0-50%
Kadar TDS Kopi 0-25%
Alat yang praktis untuk komersial, industri F&B, hobi
Brix: 0-50% TDS:0-25%
Resolution: +- 0.02C
Ambient temperature: 10-100 degrees
Measurement Time: Approximately 1.5 seconds
Minimum Sample Volume: 2 metric drops
Light Source: Yellow LED
Sample Cell: Stainless steel ring and flint glass prism
Enclosure Rating: IP65
Battery Type / Life: 9V battery for +- 5000 readings
Auto-Off: After 3 minutes of non-use
Coffee concentration with the range of 0-25%, Brix measurement with the range of 0-50% with temperature display
Calibration function and no need calibration every time; auto power off within one minute if no any operation
Provided with suction tube and spare protective cover
The function keys are simple, clear and easy to operate; high accuracy and durable
Digital screen with large display number, battery power can be displayed
1.Power On: Press key "READ" 1 second only.
2. Scale converting: press key "SCALE" 1 second to choose the unit Brixor Concentration, "S01" is for Brix %, this also can test all liquid with sugarsuch as juice, milky tea ect, range is 0-30%. "S02" is for coffeeconcentration, range is 0-25.
3. Measurement: Use suction tube takes some sample from the 1/2 deep ofthe cup (to avoid the cream or solid of the coffee), drip about 3~5 drops tothe sample plate, then cover it. Press key"READ" 1 second, meter will give the current value, ifpress key"READ" 2 seconds more, meter will make andcountdown 15 times' measurements, the reading is the average value.
4. Temperature unit converting: Press key "SCALE" more than 2 seconds,meter will display Celsius or Fahrenheit.
5. Calibration: No need calibration every time. Drip 3~5 drops distilledwater in sample plate, make sure the water and meter temperature is same andstandard temperature is 20ºC, then cover it. Press key "CAL" more than 2seconds until "CAL" flashing,press "CAL" key again, meter will display "END"and show 0.0%.
6. Power off: Meter will power off within one minute if no any operation.
7. Maintenance: Always clean and wash the sampleplate with distilled water and dry it with soft cleaning cloth, never use hard thingsto clean, otherwise it will damage the prism glass.
packing listPacking List:
1*Refractometer with retail box
1*English instruction manual